Courses

# Electromechanical Indicating Instruments To Electrical And Electronics Measurements

## 10 Questions MCQ Test Mock Test Series for Electrical Engineering (EE) GATE 2020 | Electromechanical Indicating Instruments To Electrical And Electronics Measurements

Description
This mock test of Electromechanical Indicating Instruments To Electrical And Electronics Measurements for Electrical Engineering (EE) helps you for every Electrical Engineering (EE) entrance exam. This contains 10 Multiple Choice Questions for Electrical Engineering (EE) Electromechanical Indicating Instruments To Electrical And Electronics Measurements (mcq) to study with solutions a complete question bank. The solved questions answers in this Electromechanical Indicating Instruments To Electrical And Electronics Measurements quiz give you a good mix of easy questions and tough questions. Electrical Engineering (EE) students definitely take this Electromechanical Indicating Instruments To Electrical And Electronics Measurements exercise for a better result in the exam. You can find other Electromechanical Indicating Instruments To Electrical And Electronics Measurements extra questions, long questions & short questions for Electrical Engineering (EE) on EduRev as well by searching above.
QUESTION: 1

### A moving coil instrument has the following data: Number of turns = 100 Width of the coil = 20 mm Depth of the coil = 30 mm Flux density in the gap = 0.1 Wb/m2 If the control spring constant is 2 x 10-6 N-m/ degree, then the deflection of the pointer when the coil carries a current of 10 mA is

Solution:

The deflecting torque,

At equilibrium,
Tc = Td

Thus, the deflection of the pointer  = π/6 radian.

QUESTION: 2

### A PMMC uses a :

Solution:

A PMMC uses a moving coil and the magnet is permanent.

QUESTION: 3

### A moving iron movement uses

Solution:
QUESTION: 4

A 1 mA meter movement with an internal resistance of 100 Ω is to be converted into a 0-100 mA. The value of shunt resistant required is

Solution:

QUESTION: 5

Which of the following is not a desired characteristic of shunts used in PMMC instruments?

Solution:

Shunts used in PMMC instruments are made up of manganin or constantan since they have very low temperature coefficient of resistance.

QUESTION: 6

To select the range, a multirange ammeter uses a

Solution:
QUESTION: 7

A D’Arsonval movement is

Solution:
QUESTION: 8

What is the value of multiplier resistance on the 50 V range of a dc voltmeter that uses a 500 μA meter movement with an internal resistance of 1 kΩ?

Solution:

The sensitivity of 500 μA meter movement is given by,

The value of the multiplier resistance can be obtained as

QUESTION: 9

Solution:

Higher the sensitivity of the voltmeter, lower the loading effect and vice-versa.

QUESTION: 10

Assertion (A): Gravity control is not suited for indicating instruments in general and portable instruments in particular.
Reason (R): Gravity control is used in only vertically mounted instruments.

Solution:

Gravity control is not suited for indicating instruments in general and portable instruments in particular because for using this type of control, the instruments must be mounted in level position otherwise there will be a very serious zero error. Gravity control is used in only vertically mounted instruments. Hence, reason is also a true statement but, not a correct explanation of assertion.