A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz?

## Can you answer this doubt?

People are searching for an answer to this question.
This discussion on A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz? is done on EduRev Study Group by Electrical Engineering (EE) Students. The Questions and Answers of A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz? are solved by group of students and teacher of Electrical Engineering (EE), which is also the largest student community of Electrical Engineering (EE). If the answer is not available please wait for a while and a community member will probably answer this soon. You can study other questions, MCQs, videos and tests for Electrical Engineering (EE) on EduRev and even discuss your questions like A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz? over here on EduRev! Apart from being the largest Electrical Engineering (EE) community, EduRev has the largest solved Question bank for Electrical Engineering (EE).
This discussion on A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz? is done on EduRev Study Group by Electrical Engineering (EE) Students. The Questions and Answers of A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz? are solved by group of students and teacher of Electrical Engineering (EE), which is also the largest student community of Electrical Engineering (EE). If the answer is not available please wait for a while and a community member will probably answer this soon. You can study other questions, MCQs, videos and tests for Electrical Engineering (EE) on EduRev and even discuss your questions like A 50 Volt range spring controlled, electrodynamic voltmeter having a square low scale response takes 0.05A on D.C. for full scale deflection of 900 . The control constant is 0.5 x 10-6 N-m/degree and the initial mutual inductance of the Instrument is 0.50 H. Find the true potential difference across the Instrument, when it reads 50 V at 50 Hz? over here on EduRev! Apart from being the largest Electrical Engineering (EE) community, EduRev has the largest solved Question bank for Electrical Engineering (EE).